DC (GEO-MAGNETIC, QUASI-DC) INTERFERENCE
DC refers to a static, non-varying field (0 Hz). Exceedingly low frequency fields, typically between 0 Hz and 10 Hz are referred to as quasi static or Quasi-DC or Near-DC fields.
The most common natural source of DC fields is the Earth’s magnetic field. Sometimes referred to as the geomagnetic field, this field provides a compass with the ability to indicate the direction of magnetic North.
The field strength of the Earth’s magnetic field is approximately 49 T in central Europe. Magnetic fields, including the Earth’s prefer to accumulate in ferromagnetic materials (iron, nickel or steel) rather than air. Air has a permeability of 1, while steel has a permeability of approximately 300. High-permeable metals, like mu-metal, can have permeability in the hundreds of thousands. DC magnetic lines of flux (from the Earth) will accumulate inside and near ferromagnetic materials (the steel structure of a building) and field levels in the vicinity will be elevated.
The phenomenon of elevated DC fields in structural steel can be man-made: DC welding cables can magnetize structural steel during construction, and MRIs (a source of powerful DC fields, can permanently magnetize steel in the immediate vicinity and, if physically connected, at considerable distances from the magnet. Areas of a building with magnetized steel can have DC magnetic field levels in the range of 200 T or more. As we move away from the steel structure, the field levels will decline.
Direct Current (DC) from DC source (the traction power of a railway or DC powerline) will also create a DC magnetic field. However, to the extent that the demand (load) on a DC electrical circuit will vary with time, so too will the frequency of the Quasi-DC field which it produces.
Generally, DC magnetic fields do not present an EMI threat to most electronic equipment such as office and household electronics, although in certain circumstances the ambient DC field can exceed the tolerances of sensitive instruments whose accuracy is in part based on the assumption of a stable, uniform DC field environment. And, although the earth’s magnetic field is relatively stable, there are inherent variations in both the field direction and strength that occur over time. These temporal instabilities in the earth’s magnetic field can be a source of DC interference for long-term operations, like E-Beam lithographic systems and certain electron microscope operations.
More commonly problematic are the Quasi-DC fields, which are produced by either a change in DC current (above) or by the relative movement of a ferromagnetic mass through the Earth’s DC field (below).
Quasi-DC Field Interference can disrupt the proper functioning of sensitive laboratory equipment and is a growing problem for two reasons. First, research, medical and laboratory instruments are not only proliferating, they are becoming more and more sensitive and therefore more vulnerable to EMI. Many instruments must be shielded from changes in the Earth’s magnetic field.
Second, two sources of Quasi-DC fields are growing. The first is medical and research instruments themselves; MRI’s and NMR’s emit extremely high, and occasionally ramping, DC fields, creating Quasi-DC fields.
In addition, research campuses are being built in urban areas, near public transportation. Heavy rail, light rail, buses and other large ferromagnetic bodies create quasi-DC fields which can disrupt sensitive instruments at surprisingly long distances.
This urban/internal source of DC field interference is of growing importance: the movement of large ferromagnetic bodies through the Earth’s magnetic field produces a momentary shift or perturbation in the geomagnetic field with a consequent Quasi or slowly-varying DC field. Trucks and trains outside the building add to changes caused by the movement of elevators. When the vehicles are powered by DC current (busses, trollies, rail), the changes in the current load add to the perturbation AND substantially increase the affected area. This is particularly troublesome in urban areas with subways and light rail systems.
As an offset to these risks, a careful analysis of present and future DC field conditions, during the early design stage should be performed so that (where necessary) mitigation/shielding strategies can be designed and implemented during construction to ensure each instrument’s performance at the lowest possible cost.
Compliance Engineering International is focused on providing value led engineering services across Europe. For almost twenty years we have been building our reputation as leaders in EMI mitigation and shielding services, focusing on projects in the science, research and healthcare, energy, commercial, education, transportation and government sectors.